Pocket Test Thermometer
- 1” diameter dial face
- 5” probe
- Includes protective sleeve with built-in clip
- Recalibrates for accuracy
Calibration on this unit should done with a cold bath at 32°F(0°C) for a more accurate calibration.
- Fill cup with ice and top off with cold water and stir for 30 seconds.
- Insert probe into cup at least 1″ deep for 15 seconds or until the temperature reading stabilizes. Keep the probe from touching the sides or bottom of the glass.
NOTE: If the thermometer settles at 32°F(0°C), there is no calibration needed.
- Use pliers and adjust the calibration nut on the bottom of the dial and turn until it reads the proper temperature, 32°F(0°C).